Published Jack Cochran on August 16, 2018
Quantitative VUV Spectral Deconvolution of Coeluting Methylstyrene Isomers
In a previous post, I demonstrated how styrene and some alkyl substituted analogs have unique VUV absorbance spectra. I also suggested that because of their VUV spectral uniqueness, they could be deconvoluted when they coelute in gas chromatography (GC), which is something that can’t be done with GC and mass spectrometry given that their electron ionization mass spectra are essentially identical. Spectral deconvolution can be a “who cares” exercise if compounds are coeluted unnecessarily, but with methylstyrenes, GC coelutions occur naturally.
Figure 1 is an overlay of 2-, 3-, and 4-methylstyrene VUV absorbance spectra. Hopefully you can see that each spectrum is unique. The methystyrenes coelute when using a 100% dimethylpolysiloxane GC column, so quantitative determinations for these positional isomers depend on successful deconvolution, which is accomplished across relatively wide concentration ranges when using VUV spectroscopy (Figure 2 and Table 1 and Figure 3).
If isomer analysis is important for you, take the pressure off your GC system and consider a VUV spectrometer as your detector.